The Chroma A W and A W electronic load mainframes accept the user-installable A series load modules for easy system configuration. user can perform on line voltage measurement controlled instructions via RS- C or GPIB The Chroma W and W electronic load. File, CHROMA Series, , Mainframes X, Operation Share technical documentation including product manuals, getting started guides, and.
|Published (Last):||16 April 2004|
|PDF File Size:||1.29 Mb|
|ePub File Size:||16.77 Mb|
|Price:||Free* [*Free Regsitration Required]|
There are two operation modes for Von control.
DC Electronic Load | Chroma ATE Inc.
Configuration Flexible configuration using plug-in electronic load modules to mainframes. Less Installation Space and Wiring Required.
If the input current of Load is over 40 Amps, you must use multiple terminals of connections. If input voltage is over the maximum maual low range, you must select the high range. Documentation Brake module AX Please read this document carefully before installing and commissioning the brake module!
All load modules may be configured to work synchronously, to test multiple outputs simultaneously, thus simulating real life applications. Click here to unsubscribe. Component replacement and internal adjustment can only be done by qualified service personnel.
In CV mode it is equivalent to the programming of zero voltage. Lock the module in place by use of the screwdriver see Figure usee This makes it ideal for testing.
It calls attention to an operating procedure, or the like, which, if not correctly performed or adhered to, could result in damage to or destruction of part or all of the products. Publication number E June The overcurrent OC and current error CE status register bits are set when the OC condition occurs, and will remain set till they are reset. The typical temperature coefficient is ppm. During operation the loading level mnual switched between those two load levels according to your specific setting.
If you have not done thus, Load will shut down, or fuse for remote controller in Mainframe will be broken. The wires should be large enough to limit the voltage drop to no more than 0. Asus computer user manual Lcd 3 2 e manual Solution manual for fundamentals of microelectronics by behzad razavi victory vegas 8 ball service manual Canon zr80 manual pdf Honda hrv manual indonesia Kdxr service manual Canon cr service manual Severn glocon control valve manual Bissell prodry tlr power roller manual Rabbit service manual pdf Hsm 2 manual Allen bradley plc 5 programming manual Samsung manual Montgomery ward microwave oven manuals Audi a6 user manual Dell photo aio printer manual Dell openmanage manual chevy truck repair manual Casio salarms manual.
Avoid touching the connector and the circuit board. It is designed for applications in research and development, production, and incoming inspection.
Measurement and Load Cables 2.
Programmable DC Electronic Load Series Operation & Programming Manual – PDF
Programmable dynamic loading with speed up to 20KHz. If the factory set switch on this label does chrima correspond to your nominal line voltage, turn the Mainframe power off, and disconnect the power cord. Daker DK 1, 2, 3 kva. In local mode any value can be set to a module from the keypad.
Channel number is automatically assigned to each channel: The nominal output power is More information. The A load modules operate in constant current, constant voltage, constant power or constant resistance to satisfy a wide range of test requirements. The unit has been factory set to the correct line voltage.
Programmable DC Electronic Load 6310 Series Operation & Programming Manual
The voltage range selection of CC mode is in configuration setting. In the chrima state, the keys on the front panel will have no effect. Selective voltage and current ranges. The system is configured by plugging the user selectable load modules into the system mainframe.
This manual is the best copy we could find; it may be incomplete or contain dated information.
Operation & Programming
The mainframe will scan module type at power-on, and memorize it for channel setting. Local operation on front panel keypad.
nanual Class B devices have a Federal Communications. Control by Key means pressing SHORT and holding it to enable short circuit, and releasing it to return to normal operation.
One is 3614 rise rate, and another for fall rate. You can start sinking current when setting is load ON, and the input voltage of the module is over Von voltage, but stop sinking when load OFF, or the input voltage below Von voltage. In remote mode programmed value cannot be over boundary.