KiWAVE™ is a trademark of Keithley Instruments, Inc. All Keithley Instruments product names are .. Model Arbitrary Waveform Generator User’s Manual. Items are supplied with manuals, accessories and typically a full no-quibble 2 The exceptional signal quality of the Model is a result of its high resolution, . Refer to the Keithley Model Arbitrary Waveform Generator User’s Manual if it is not clear how to specify the waveforms. The figure below shows an.
|Published (Last):||9 February 2018|
|PDF File Size:||16.22 Mb|
|ePub File Size:||4.92 Mb|
|Price:||Free* [*Free Regsitration Required]|
KEITHLEY USER MANUAL Pdf Download.
When you first receive your instrument, it is secured. Press the soft keys corresponding with frequency sweep Select a start frequency and a stop frequency, or select a center frequency and a frequency span: SCPI command summary Pattern output operation.
Page 42 Section 4: Page 78 For example, the domain “www. Also See for User manual – pages. Page of Go. Waveform Output Operations The default source is internal with a default frequency of Hz the acceptable range is 2 mHz to 20 kHz.
Keithley 3390 Function Generator, Frequency Generator 3390 Data Sheet
A single generated frequency provides all of the output frequencies. Before you adjust the instrument, unsecure it by inputting the correct security code. Setting up a pulse waveform To Comments to this Manuals Your Name.
Introduction A sine waveform is a uniform waveform with a constant frequency and amplitude. Introduction Error messages are issued when an incorrect command is sent to the Keithley Instruments Model Arbitrary Waveform Generator.
Page 96 Appendix A: General functions and settings The default source is internal with a default frequency of 10 Hz the acceptable range is 2 mHz to 20 kHz.
Refer to Section 4, Setup Basics for Insert the GPIB interface card into the interface slot on the waveform generator’s rear panel. Set the IP address to Setting Voltage Auto Ranging Section 4: The IEEE end-or-identify EOI message is interpreted as a new line character and can be used to terminate a command string in place of a new line character.
Also, the Model expands the number of points to fill the waveform memory; less than 16, 16K points will automatically generate a waveform with 16, points. Topic Page Rear panel description To print the manual completely, please, download it. To maintain protection from electric shock and fire, replacement components in mains circuits – including the power transformer, test leads, and input jacks – must be purchased from Keithley Instruments.
Menu operation manuql keys Page 5 Any unauthorized reproduction, photocopy, or use the information herein, in whole or in part, without the prior written approval of Keithley Instruments, Inc. Waveform Output Operations Figure manuwl Topic Page Rear panel description System Operations To store a state from a remote interface: Command errors Error code Description A discrete parameter was received when it was not Character data not allowed expected.
Operation position Figure Connect And Power The Instrument Follow the procedure below to connect the Model to line power and turn on the instrument.
Table of Contents Add to my manuals Add. Ventilation Failure to use a grounded outlet may result in personal injury or death due to electric shock. Controlling the system sound Section 5 Waveform Output Operations In this section: General information Contact information If you have any questions after reviewing this information, please contact your local Keithley Instruments representative or call one of our applications engineers at KEITHLEY within the U.
This cable should be used when generating a pattern output function. Index Numerics F 10 MHz connectors Waveform Output Operations 5. Storing the instrument state The following text provides detailed information regarding handle adjustment and removal. Section 3 Rear Panel In this section: Internal Timebase Verification Section 1: All Keithley Instruments product names are trademarks or registered trademarks of Keithley Instruments, Inc.
Failure to use a grounded outlet may result in personal injury or death due to electric shock. This command enables the output immediately.
Date Name and Telephone No. Operation handle position Position